Cold test measurements on components of the 1 MW, 140 GHz, CW gyrotron for the stellarator Wendelstein 7-X
For the development of a 1 MW, 140 GHz gyrotron for CW operation which will be installed at the stellarator facility Wendelstein 7-X at IPP Greifswald, a collaboration between different European research institutes and an industrial company has been established. In order to prove the proper functioning of the millimeter wave components installed in the gyrotron - such as the cavity, the waveguide taper and the quasioptical mode converter - these components should be cold tested, preferably before installation. However, due to lack of time as well as long delivery times, this was not possible. Therefore, two units of the quasioptical mode converter and the cavity were fabricated with identical geometry, one of those being used for measurements on the low power test device. To perform these cold tests for tapers and mode converters, the gyrotron cavity output mode has to be simulated. This means that a high order rotating mode (TE28,8 mode) must be generated at low power. This can be achieved by means of a mode generator consisting of two mirrors and a coaxial cavity with a perforated outer wall. Before applying the mode generator to the components, its proper behavior and the accurate alignment of the system must be verified either by radiation pattern measurements or k-spectrometer measurements. As the coupling through the holes of the k-spectrometer is extremely low, a special vector network analyzer with a dynamic range of at least 100 dB had to be developed. This has been achieved by integration of a phase locked backward-wave oscillator with a line width of 100 Hz and an output power of 10 mW. A non-destructive measurement of the resonance frequency and the quality factor of the cavity does not seem possible. The second cavity will be prepared for the cold measurement by drilling a small radial hole into its wall in the plane of the field maximum. This hole is then used for the input coupling. The accuracy required for this hole is rather critical. The coupling coefficient must be high for sufficient excitation of the rf field, but on the other hand it must neither change the frequency nor the quality factor strongly. The transmission is measured by a probe at the output of the uptaper. (C) 2001 Elsevier Science B.V. All rights reserved.