Principle of ferroelectric domain imaging using atomic force microscope
The contrast mechanisms of domain imaging experiments assisted by atomic force microscope (AFM) have been investigated by model experiments on nonpiezoelectric (silicon oxide) and piezoelectric [Pb(Zr,Ti)O-3] thin films. The first step was to identify the electrostatic charge effects between the tip, the cantilever, and the sample surface. The second step was to explore the tip-sample piezoelectric force interaction. The static deflection of the cantilever was measured as a function of dc bias voltage (V-dc) applied to the bottom electrode (n-type Si wafers) for noncontact and contact modes. In addition, a small ac voltage (V-ac sin omegat) was applied to the tip to measure the amplitude (A(omega)) and phase (Phi (omega)) of the first harmonic (omega) signal as a function of V-dc. By changing from the noncontact to the contact mode, a repulsive contribution to the static deflection was found in addition to the attractive one and a 180 degrees phase shift in Phi (omega) was observed. These results imply that in the contact mode the cantilever buckling is induced by the capacitive force between the cantilever and the sample surface. This interaction adds to the tip-sample piezoelectric interaction thereby overlapping the obtained tip vibration signal. Therefore, the antiparallel ferroelectric domain images obtained at zero dc bias voltage will show a variation in A(omega) but a negligible one in Phi (omega). The capacitive force contribution to the tip vibration signal was further verified in piezoelectric hysteresis loop measurement assisted by the AFM. The observed vertical offset of the loops was explained by the contact potential difference between the cantilever and the bottom electrode. The shape of the curve could be explained by the capacitive force interaction combined with the tip-sample piezoelectric interaction. The experimental results obtained in this study support the interpretation of the cantilever-sample capacitive force contribution to the tip vibration signal in ferroelectric domain imaging experiments using AFM as a probing tool. The use of a large area top electrode between the tip and the sample resulted in the elimination of the electrostatic cantilever-sample interaction with negligible degradation of the domain contrast. This method proved to be successful because the cantilever-sample interaction was hardly detected and only the tip-sample interaction was observed. (C) 2001 American Institute of Physics.
- View record in Web of Science
Hong, S Swiss Fed Inst Technol, Ceram Lab, CH-1015 Lausanne, Switzerland Samsung Adv Inst Technol, Nano Syst Lab, Suwon, South Korea Korea Adv Inst Sci & Technol, Elect & Opt Mat Lab, Taejon 305701, South Korea Creat Res Initiat, Suwon, South Korea Samsung Adv Inst Technol, MEMS Lab, Suwon, South Korea Swiss Fed Inst Technol, Ceram Lab, CH-1015 Lausanne, Switzerland
Cited References Count:12
Record created on 2006-08-21, modified on 2016-08-08